- Short report
- Open Access
Extraction of total nucleic acids from bacterial isolates using the bioMérieux NucliSENS easyMAG total nucleic acid extractor
© The Author(s) 2016
- Received: 17 May 2016
- Accepted: 13 September 2016
- Published: 22 September 2016
The BioMerieux NucliSENS easyMAG total nucleic acid extractor was evaluated for use on bacterial isolates in the clinical microbiology laboratory. Forty eight isolates were extracted, yielding quantifiable amounts of DNA for all isolates. The easyMAG is appropriate for DNA extraction from bacterial isolates and will be incorporated in the clinical laboratory.
- DNA extraction
- 16S rRNA sequencing
- Bacterial sequencing
The number of molecular methods and their importance in the clinical microbiology laboratory is rapidly increasing. Nucleic acid extraction from bacterial isolates is an important first step in several molecular test methodologies, including 16S rRNA sequencing for bacterial identification, whole genome sequencing, and other genotypic methods. While some clinical microbiology testing platforms have integrated nucleic acid extraction, others require separate nucleic acid extraction before testing may begin. Laboratory quality control has recently come under renewed scrutiny with the implementation of individual quality control plan (IQCP) requirements and as part of this, it is critically important to identify steps in a testing process that may result in testing failure or testing delays.
Inadequate nucleic acid extraction may result in an insufficient amount of nucleic acids, which can result in wasted reagents and delayed results. Manual extraction methods have traditionally been used, but these methods require extensive hands-on time and frequently have many steps, which increases the opportunity for method error or sample contamination . The NucliSENS easyMAG total nucleic acid extractor (bioMérieux, Marcy l’Etoile, France) is a platform capable of total nucleic acid extraction from various sources, including blood, stool, and urine. The easyMAG extracts nucleic acids using magnetic silica beads. EasyMAG extraction of clinical samples such as respiratory samples, stool, and blood followed by molecular detection of viral and bacterial pathogens has been reported [2–4]. While the easyMAG has been used on a variety of clinical specimen types, the extractor does not include pre-set protocols for bacterial extraction. Although a small number of published studies have utilized the easyMag extractor to extract nucleic acids from bacterial isolates, these have typically been limited to a small range of species . For the easyMAg to be utilized as the primary bacterial nucleic acid method in a clinical microbiology laboratory, the extractor must be capable of extracting nucleic acids from a wide variety of bacterial species. To determine the suitability of the easyMAG for routine bacterial nucleic acid extraction, the current study sought to examine the ability of the easyMag extractor to extract nucleic acids from a wide variety of bacterial isolates.
Bacterial species included in validation
Non-Enterobacteriaceae gram-negative organisms
Escherichia coli (3)
Pseudomonas aeruginosa (2)
Proteus mirabilis (2)
For each of the four groups, the two isolates with the lowest nucleic acid concentrations were submitted for 16S rRNA sequencing. When these two isolates were from the same genus, the isolate with the next lowest nucleic acid concentration was submitted, ensuring more diverse representation of each group. Sequencing results were compared to the ATCC identification of the organism to ensure correct identification. Four additional isolates yielding higher extracted DNA concentrations (7.8–21.6 ng/µL) were also sequenced.
Concentration and A260/A280 of extracted total nucleic acid
DNA concentration (ng/µL)
In-house nucleic acid extraction can decrease turn-around-time of downstream testing and minimizes the risks associated with sending potential pathogens to other laboratories. Coupling laboratory-performed nucleic acid extraction with nucleic acid quantification allows laboratories to perform additional manipulation on isolates that initially do not yield adequate amounts of nucleic acid. Identification of these isolates before downstream testing can prevent unsuccessful testing, save reagents and testing time, and decrease overall time to test result. While this was not required in the current study, lysis can be performed manually rather than on the instrument to extend lysis time and additional physical lysis steps such as grinding can also be added. Additional studies are necessary to develop protocols for nucleic acid extraction from acid-fast bacilli and fungi. The efficacy, flexibility, and ease-of-use of the bioMérieux NucliSENS easyMAG total nucleic acid extractor make it suitable for use in the routine clinical microbiology laboratory.
EAP and JEM made substantial contributions to the study design, acquisition and analysis of the data. Both authors read and approved the final manuscript.
The authors declare that they have no competing interests.
No outside funding sources.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Loens K, Bergs K, Ursi D, Goossens H, Leven M. Evaluation of NucliSens easyMAG for automated nucleic acid extraction from various clinical specimens. J Clin Microbiol. 2007;45(2):421–5.View ArticlePubMedGoogle Scholar
- Esona MD, McDonald S, Kamili S, Kerin T, Gautam R, Bowen MD. Comparative evaluation of commercially available manual and automated nucleic acid extraction methods for rotavirus RNA detection in stools. J Virol Methods. 2013;194(1–2):242–9.View ArticlePubMedPubMed CentralGoogle Scholar
- Kalina WV, Douglas CE, Coyne SR, Minogue TD. Comparative assessment of automated nucleic acid sample extraction equipment for biothreat agents. J Clin Microbiol. 2014;52(4):1232–4.View ArticlePubMedPubMed CentralGoogle Scholar
- Manji R, Zheng X, Patel A, Kowerska M, Vossinas M, Drain A, Todd KM, Lenny N, DeVincenzo JP, Ginocchio CC. Multi-center evaluation of the adenovirus R-gene US assay for the detection of adenovirus in respiratory samples. J Clin Virol. 2014;60(2):90–5.View ArticlePubMedGoogle Scholar
- Monteiro J, Widen RH, Pignatari AC, Kubasek C, Silbert S. Rapid detection of carbapenemase genes by multiplex real-time PCR. J Antimicrob Chemother. 2012;67(4):906–9.View ArticlePubMedGoogle Scholar
- Manchester KL. Value of A260/A280 ratios for measurement of purity of nucleic acids. Biotechniques. 1995;19(2):208–10.PubMedGoogle Scholar