- Open Access
Verification of the ProPneumo-1 assay for the simultaneous detection of Mycoplasma pneumoniae and Chlamydophila pneumoniae in clinical respiratory specimens
Annals of Clinical Microbiology and Antimicrobials volume 8, Article number: 10 (2009)
Mycoplasma pneumoniae and Chlamydophila pneumoniae are major causes of lower and upper respiratory infections that are difficult to diagnose using conventional methods such as culture. The ProPneumo-1 (Prodesse, Waukesha, WI) assay is a commercial multiplex real-time PCR assay for the simultaneous detection of M. pneumoniae and/or C. pneumoniae DNA in clinical respiratory samples.
The aim of this study was to evaluate the sensitivity and specificity of the ProPneumo-1, a newly developed commercial multiplex real-time PCR assay.
A total of 146 clinical respiratory specimens, collected from 1997 to 2007, suspected of C. pneumoniae or M. pneumoniae infections were tested retrospectively. Nucleic acid was extracted using an automated NucliSense easyMag (bioMerieux, Netherlands). We used a "Home-brew" monoplex real-time assay as the reference method for the analysis of C. pneumoniae and culture as the reference method for the analysis of M. pneumoniae. For discordant analysis specimens were re-tested using another commercial multiplex PCR, the PneumoBacter-1 assay (Seegene, Korea).
Following discordant analysis, the sensitivity of the ProPneumo-1 assay for pathogens, C. pneumoniae or M. pneumoniae, was 100%. The specificity of the ProPneumo-1 assay, however, was 100% for C. pneumoniae and 98% for M. pneumoniae. The limits of detection were 1 genome equivalent (Geq) per reaction for pathogens, M. pneumoniae and C. pneumoniae. Due to the multipex format of the ProPneumo-1 assay, we identified 5 additional positive specimens, 2 C. pneumoniae in the M. pneumoniae-negative pool and 3 M. pneumoniae in the C. pneumoniae-negative pool.
The ProPneumo-1 assay is a rapid, sensitive and effective method for the simultaneous detection of M. pneumoniae and C. pneumoniae directly in respiratory specimens.
Mycoplasma pneumoniae and Chlamydophila pneumoniae are major causes of lower and upper respiratory tract infection in adults and children [1–3] with high incidence occurring among schoolchildren [4, 5]. Both pathogens are difficult to diagnose using conventional methods such as culture and serology. Furthermore, respiratory infection may be caused by one of several bacterial or viral pathogens that share similar symptoms and clinical features. To overcome difficulties in the detection of these pathogens, PCR assays have been developed for the rapid detection and identification of pathogens directly in respiratory samples . PCR assays are widely used in the clinical arena with real-time PCR slowly replacing conventional PCR assays. Several "home-brew" single target (monoplex) assays have been utilized for the detection of numerous pathogens, including M. pneumoniae [7–9]. However, there are multiple regulatory issues with the use of "home-brew" assays and controversies exist as to their use in a diagnostic laboratory setting. As commercial assays become more available, there is a growing movement to replace "home-brew" assays with commercially available products [10, 11]. As well, the monoplex PCR format is being replaced by a multiplex one. The aim of this study was to evaluate the analytical sensitivity and the diagnostic specificity and sensitivity of the ProPneumo-1 assay, a newly developed commercial multiplex real-time PCR assay for the simultaneous detection of M. pneumoniae and/or C. pneumoniae DNA in clinical respiratory specimens.
Specimen collection and analysis by the reference method
Historically, suspected C. pneumoniae infections were tested using a "home-brew" single target monoplex real-time PCR assay  and suspected M. pneumoniae infections was tested using culture. As shown in Table 1, we examined a total of 146 clinical respiratory specimens; of which 58 were previously tested for C. pneumoniae using a "home-brew" real-time PCR and 88 were previously tested for M. pneumoniae using culture. Specimens were collected in Ontario, between 1997 and 2007, from patients with suspected C. pneumoniae or M. pneumoniae infections. The types of specimen submitted for testing at the Central Public Health Laboratory in Toronto, Ontario, including patient's sex and age are described in Table 2 for M. pneumoniae, and 2B for C. pneumoniae. Specimens and extracted nucleic acid were stored at -80°C until use.
DNA for "home-brew"monoplex real-time PCR was extracted from specimens using the MagNA Pure LC (Roche, Mannheim, Germany) as per manufacturer's instructions. Specimens were tested using a "home-brew" single target monoplex real-time PCR assay . Real-time amplification was carried out using the Roche Light Cycler (Roche, Mannheim, Germany), as previously published [12, 13].
Clinical specimens from patients with suspected M. pneumoniae infection were cultured in PPLO medium containing thallium acetate, horse serum, yeast dialysate and supplemented with amphotericin B (0.5 mg/ml), penicillin G (100,000 u/ml) and nystatin (50,000 u/ml). Biphasic culture flasks were inoculated with specimens, incubated at 37°C and inspected daily for 4 weeks. The organism was identified based on typical colonial morphology on the agar medium and the change in the broth color from red to orange then to yellow in the absence of turbidity of the broth.
ProPneumo-1 and PneumoBacter-1 assays
Nucleic acid, for analysis by the ProPneumo-1 (Prodesse, Waukesha, WI) or the PneumoBacter-1 (Seegene, Korea) assay, was extracted using the easyMag NucliSense magnetic extraction (bioMerieux, Netherlands) . Each nucleic acid extract was amplified by both, the PromPneumo-1 and PneumoBacter-1 multiplex PCR assays. The ProPneumo-1 assay was performed as per manufacturer's instructions. Amplification and detection of the ProPneumo-1 test was performed on the ABI 7500 (Applied Biosystems, Foster City, CA). For discordant analysis all specimens, including concordant specimens, were re-tested using the PneumoBacter-1 assay (Seegene, Korea). PneumoBacter-1 is an agarose gel-based assay, amplification was performed on the iCycler (Bio-Rad, Milpitas, CA) and amplicons were imaged, following gel electrophoresis, as per manufacturer's protocols.
Experimental controls: external, internal and extraction controls
M. pneumoniae and C. pneumoniae quantified DNA controls (Advanced Biotechnologies Inc., Columbia, MD) were included in each assay as external positive controls. For internal DNA control and to monitor for potential PCR inhibitors, exogenous DNA was spiked into each specimen. To monitor for the integrity of extracted nucleic acid, all specimens were also tested for the presence of the housekeeping gene, human glyceraldehyde phosphate dehydrogenase (GAPDH) (Applied Biosystems. Foster City, CA). Real-time amplification and detection of the GAPDH gene sequence was performed on the ABI 7900HT (Applied Biosystems, Foster City, CA), as previously described [15, 16]. This test was required to ascertain the presence of nucleic acid in specimens, which tested negative for C. pneumoniae or M. pneumoniae. For negative controls, target nucleic acid and specimens were replaced by PCR grade water in each PCR assay or nucleic acid extraction run, respectively.
Limits of detection
The limits of detection for the assays described in this study, ProPneumo-1 and PneumoBacter-1, were determined using 10 fold serial dilutions of quantified M. pneumoniae and C. pneumoniae DNA (Advanced Biotechnologies Inc., Columbia, MD) in PCR-grade water. Data analysis was carried out using SPSS 15.0 (SSPS Inc., Chicago, IL). Turn-around-times per assay were estimated from a review of laboratory procedures and include time from specimen reception to results reporting.
The following definitions were used for analysis. Concordant specimen: agreement between "home-brew" assay result and Prodesse ProPneumo-1 assay result. Discordant specimen: disagreement between "home-brew" assay result and Prodesse ProPneumo-1 assay result. Resolved specimen: agreement in result, either positive or negative between any two of the following assays; 1) "home-brew" or culture, 2) ProPneumo-1, and 3) PneumoBacter-1.
Table 1 summarizes the results obtained for the evaluation of the ProPneumo-1 real time multiplex PCR assay. Overviews of the results obtained for subsets of M. pneu moniae and C. pneumoniae patients are presented in Table 2 and Table 3, respectively. Among patients who tested positive for either pathogen, the ratio of male to female is approximately the same and, with the majority of positive patients being under the age of 20 years. Table 2 also displays the results obtained for each of the three tests used to identify positive M. pneumoniae (A) or C. pneumoniae (B) patients. Using a monoplex "home-brew" real-time PCR as a reference in the analysis of C. pneumoniae and culture as a reference in the analysis of M. pneum oniae, 14 and 41 specimens tested positive for C. pneumoniae and M. pneumoniae, respectively by the ProPneumo-1 assay. As shown in Table 1, 11 patients tested positive for C. pneumoniae by the "home-brew" monoplex assay and 42 patients tested positive for M. pneumoniae by culture. Similarly, 48 and 43 specimens were defined as negative in the ProPneumo-1 assay, for M. pneumoniae and C. pneumoniae, respectively. Although, 42 positive M. pneumoniae specimens were identified by culture and 41 by the ProPneumo-1assay, the identity of 8 positive specimens however, varies between the two methods (see Table 2). Curiously, both assays the ProPneumo-1 and PneumoBacter-1 detected 5 additional positive specimens, 3 for M. pneum oniae and 2 for C. pneumoniae in the C. pneumoniae-negative and M. pneumoniae-negative pool, respectively.
The limit of detection of the ProPneumo-1 assay as determined by Probit regression (95% probability) was estimated as follows: 1 genome equivalent (Geq)/reaction for M. pneumoniae and 1 Geq/reaction for C. pneumoniae. The limit of detection for PneumoBacter-1 was 1 Geq/reaction for M. pneumoniae and 1 Geq/reaction for C. pneumoniae. The turn-around-times, from specimen reception to reporting, are as follows: ProPneumo-1, 6 hours; "home-brew" monoplex PCR, 10 hours; PneumoBacter-1, 12 hours, and Mycoplasma culture, 4 weeks.
Diagnostic specificity and sensitivity
To determine the diagnostic specificity and sensitivity of the ProPneumo-1 assay, results were scored against the reference method ["home-brew" monoplex real-time PCR for C. pneumoniae (see Table 4) and culture for M. pneumoniae (Table 5)]. To resolve discordant results, specimens were re-tested using the multiplex PneumoBacter-1 method (Tables 4 and Table 5).
The ProPneumo-1 assay detected 14 positive C. pneumoniae specimens compared to 11 in the monoplex "home-brew" assay, of which 1 specimen was false positive. Similarly the ProPneumo-1 detected 43 negative C. pneumoniae specimens compared to 47 in the monoplex "home-brew" assay, of which 2 specimens were false negative (Table 1 and Table 3). Prior to discordant analysis, the sensitivity and specificity of the ProPneumo-1 assay for C. pneumoniae was 91% and 96%, respectively (Table 4).
In the case of M. pneumoniae, ProPneumo-1 and culture detected 42 and 41 positive specimens, respectively. However, 4 specimens scored positive by the culture method (Table 1 and Table 2) and negative by the PCR method. Likely, due to repeated freeze-thaw cycles and the long storage period at -80°C (up to 10 years), these 4 specimens are devoid of intact bacterial DNA and yielded negative results in both PCR tests. Similarly, one specimen appeared false positive by the ProPneumo-1 assay. We suggest that this specimen is likely a true positive which was not detected by the PneumoBacter-1 assay, likely due to its gel-based format.
Prior to discordant analysis the sensitivity and specificity of the ProPneumo-1 assay for M. pneumoniae was 95% and 98%, respectively (Table 5). Following discordant analysis by the PneumoBacter-1 multiplex assay and classification of specimens as per the definitions described above, the sensitivity and specificity of the ProPneumo-1 assay for C. pneumoniae was 100% and 100%, respectively (Table 4). The sensitivity and specificity of the ProPneumo-1 assay for M. pneumoniae, however, was 100% and 98%, respectively (Table 5). Consistent with its multiplex format, the ProPneumo-1 assay detected 5 additional positive specimens, 2 C. pneumoniae in the M. pneumoniae-negative pool and 3 M. pneumoniae in the C. pneumoniae-negative pool.
Early work in molecular diagnostics utilized technologies that were considered to be site-developed or "home-brew" in nature. Although these assays were considerable advances at their time, they were often plagued by requirements for up-front research and development, quality assurance, and laboratory accreditation issues [10, 11]. As a result, past molecular diagnostics for the clinical microbiology laboratory were often restricted to laboratories that had the monetary and human resources to meet these demands. In the last five years however, commercial products such as those identified in this manuscript have allowed for the implementation of molecular diagnostics without the up-front requirement for research and development as well as simplified quality assurance steps.
The ProPneumo-1 assay demonstrated several strengths as a commercial product for use in a wider range of clinical microbiology laboratory settings outside of the reference or highly resourced setting. First, this assay is a rapid, sensitive and specific method that allows for the diagnosis of C. pneumoniae and M. pneumoniae in a variety of clinical specimens. Second, the fact that the Pro-Pneumo-1 assay is a commercial product allows for simpler implementation in the diagnostic laboratory, and continued validation (e.g. quality assurance) when compared to "home-brew" assays [10, 11]. Third, the ProPneumo-1 assay is a multiplex assay, which allows for multiple primer combinations in one reaction, instead of multiple monoplex reactions. This not only makes the assay easier to perform in a laboratory but also has the potential of utilizing less patient specimen . Indeed, using the ProPneumo-1 multiplex assay we detected 5 additional positive specimens, 2 C. pneumoniae in the M. pneumoniae-negative pool and 3 M. pneumoniae in the C. pneumoniae-negative pool.
There are several limitations to this study including its retrospective nature. A further limitation of this study is the relatively low number of C. pneumonia-positive specimens analyzed. However, the number of specimens analyzed represents the total complement of specimens available from this region over a period of ten years. The authors hope that further prospective analysis will allow for the further characterization of this commercial assay.
Several studies have evaluated the use of multiplex PCR for the detection of M. pneumoniae and C. pneumoniae in clinical specimens [18–20]. These studies have used a variety of assays, patient populations and sample types, thus proper validation and standardization are often lacking. These factors make it difficult to compare different studies although the finding has been generally consistent. A study by Khanna and colleagues  described the Pneumoplex assays (standard and real-time PCR) that can detect 5 pathogens, including M. pneumoniae and C. pneumoniae, in a single reaction. The diagnostic sensitivity of the Pneumoplex assay was 100% and was defined by spiking negative BAL specimens with bacterial DNA. The diagnostic specificity and sensitivity of the ProPneumo-1 assay, on the other hand, were determined in this study directly from patient's specimens. Another study by Ginevra and colleagues  described the triplex Chlamylege assay, for the detection of Legionella, M. pneumoniae and C. pneumoniae. The diagnostic specificity of the Chlamylege assay was also 100% and it was defined by sequencing the DNA of discordant specimens. Compared to our ProPneumo-1 study, fewer positive specimens were used for the evaluation of the Chlamylege assay (2 C. pneumoniae and 9 M. pneumoniae). However, the lack of an of appropriate "gold standard" for the quantitative analysis of intracellular pathogens makes comparisons of the sensitivities of these different assays difficult.
In conclusion, the ProPneumo-1 real-time multiplex PCR assay is a sensitive, specific, convenient and reliable tool for the detection of atypical respiratory infection pathogens directly from respiratory tract specimens. The authors believe that this product can be effectively implemented in a wide variety of clinical microbiology settings outside of reference and highly resourced laboratory settings.
Stralin K, Tornqvist E, Kaltoft MS, Olcen P, Holmberg H: Etiologic diagnosis of adult bacterial pneumonia by culture and PCR applied to respiratory tract samples. J Clin Microbiol. 2006, 44: 643-645. 10.1128/JCM.44.2.643-645.2006
Loens K, Ursi D, Goossens H, Ieven M: Molecular diagnosis of Mycoplasma pneumoniae respiratory tract infections. J Clin Microbiol. 2003, 41: 4915-4923. 10.1128/JCM.41.11.4915-4923.2003
Waites KB, Talkington DF: Mycoplasma pneumoniae and its role as a human pathogen. Clin Microbiol Rev. 2004, 17: 697-728. 10.1128/CMR.17.4.697-728.2004
Heath PT: Epidemiology and bacteriology of bacterial pneumonias. Paediatr Respir Rev. 2000, 1: 4-7. 10.1053/prrv.2000.0001
McIntosh K: Community-acquired pneumonia in children. N Engl J Med. 2002, 346: 429-437. 10.1056/NEJMra011994
Daxboeck F, Krause R, Wenisch C: Laboratory diagnosis of Mycoplasma pneumoniae infection. Clin Microbiol Infect. 2003, 9: 263-273. 10.1046/j.1469-0691.2003.00590.x
Loens K, Ieven M, Ursi D, Beck T, Overdijk M, Sillekens P, Goossens H: Detection of Mycoplasma pneumoniae by real-time nucleic acid sequence-based amplification. J Clin Microbiol. 2003, 41: 4448-4450. 10.1128/JCM.41.9.4448-4450.2003
Winchell JM, Thurman KA, Mitchell SL, Thacker WL, Fields BS: Evaluation of three real-time PCR assays for detection of Mycoplasma pneumoniae in an outbreak investigation. J Clin Microbiol. 2008, 46: 3116-3118. 10.1128/JCM.00440-08
Dumke R, Schurwanz N, Lenz M, Schuppler M, Luck C, Jacobs E: Sensitive detection of Mycoplasma pneumoniae in human respiratory tract samples by optimized real-time PCR approach. J Clin Microbiol. 2007, 45: 2726-2730. 10.1128/JCM.00321-07
Gammie AJ: For the proposition: for the diagnosis of viral infections, commercial assays provide more reliable results than do in-house assays. Rev Med Virol. 2008, 18: 73-76. 10.1002/rmv.564
James V: Against the proposition: for the diagnosis of viral infections, commercial assays provide more reliable results than do in-house assays. Rev Med Virol. 2008, 18: 77-80. 10.1002/rmv.562
Tondella ML, Talkington DF, Holloway BP, Dowell SF, Cowley K, Soriano-Gabarro M, Elkind MS, Fields BS: Development and evaluation of real-time PCR-based fluorescence assays for detection of Chlamydia pneumoniae. J Clin Microbiol. 2002, 40: 575-583. 10.1128/JCM.40.2.575-583.2002
Ursi D, Dirven K, Loens K, Ieven M, Goossens H: Detection of Mycoplasma pneumoniae in respiratory samples by real-time PCR using an inhibition control. J Microbiol Methods. 2003, 55: 149-153. 10.1016/S0167-7012(03)00131-3
Loens K, Ursi D, Goossens H, Ieven M: Evaluation of the NucliSens miniMAG RNA extraction and real-time NASBA applications for the detection of Mycoplasma pneumoniae and Chlamydophila pneumoniae in throat swabs. J Microbiol Methods. 2008, 72: 217-219. 10.1016/j.mimet.2007.11.008
Drews SJ, Blair J, Lombos E, DeLima C, Burton L, Mazzulli T, Low DE: Use of the Seeplex RV Detection kit for surveillance of respiratory viral outbreaks in Toronto, Ontario, Canada. Ann Clin Lab Sci. 2008, 38: 376-379.
Eshaghi A, Blair J, Burton L, Choi KW, De LC, Duncan C, Guyard C, Higgins R, Lombos E, Low DE, Mazzulli T, Drews SJ: Characterization of an influenza A and influenza B co-infection of a patient in a long-term care facility with co-circulating influenza A and influenza B. Int J Infect Dis. 2008.
Brice G, Drews SJ, Low DE: Respiratory virus panels for global surveillance of emerging infectious diseases. J Clin Virol. 2007, 40: S58-S60. 10.1016/S1386-6532(07)70014-8
Boman J, Gaydos CA, Quinn TC: Molecular diagnosis of Chlamydia pneumoniae by infection. J Clin Microbiol. 1999, 37: 3791-3799.
Khanna M, Fan J, Pehler-Harrington K, Waters C, Douglass P, Stallock J, Kehl S, Henrickson KJ: The pneumoplex assays, a multiplex PCR-enzyme hybridization assay that allows simultaneous detection of five organisms, Mycoplasma pneumoniae, Chlamydia (Chlamydophila) pneumoniae, Legionella micadadei, and Bordetella pertussis, and its real-time counterpart. J Clin Microbiol. 2005, 43: 565-571. 10.1128/JCM.43.2.565-571.2005
Ginevra C, Baranger C, Ros A, Mory O, Stephan J-L, Freymuth F, Joannes M, Pozzeto B, Grattard F: Development and evaluation of chlamylege, a new commercial test allowing simultaneous detection and identification of Legionella, Chalmydophila pneumoniae, and Mycoplasma pne umoniae in clinical respiratory specimens by multiplex PCR. J Clin Microbiol. 2005, 43: 3247-3254. 10.1128/JCM.43.7.3247-3254.2005
We thank Raymond Poon and Mary Morrison for help and technical expertise.
The authors declare that they have no competing interests.
RRH undertook data analysis, assisted in validation study planning and drafted a manuscript. EL and PT participated in validation study planning and work. KR, AM, and SB undertook specimen collection and reference method analysis. FJ assisted in planning of experiments. SJD planned and coordinated study and drafted manuscript.
About this article
Cite this article
Higgins, R.R., Lombos, E., Tang, P. et al. Verification of the ProPneumo-1 assay for the simultaneous detection of Mycoplasma pneumoniae and Chlamydophila pneumoniae in clinical respiratory specimens. Ann Clin Microbiol Antimicrob 8, 10 (2009). https://doi.org/10.1186/1476-0711-8-10
- Mycoplasma Pneumoniae
- Diagnostic Specificity
- Clinical Microbiology Laboratory
- Chlamydophila Pneumoniae
- Respiratory Tract Specimen